Mobile Erythemal Radiometer
The XD-45-ERYC-4 irradiance detector is specially designed for the measurement of the Erythemal effective UV radiation in tanning equipment. Gigahertz-Optik’s X14 optometer enables health and safety executives to do on-site measurements at the tanning salon conforming to published safety standards.
Measurement of the UV-C Limit Value
The relevant German UVSV (Verordnung zum Schutz vor schädlichen Wirkungen künstlicher ultravioletter Strahlung) specifies the limit values for Erythemal effective irradiance at 0.3 W/m² within the wavelength range from 250 nm to 400 nm. From 200 nm to 280 nm the irradiance should not exceed 0.003 W/m². The XD-45-ERYC-4 detector is set-up with a third UV-C sensor to measure any existing UV-C radiation at 254 nm with low uncertainty.
Cosine Corrected Field of View
The angular response of the XD-45-ERYC-4 detector is cosine corrected.
Calibration of the detector responsivity is performed by the Gigahertz-Optik GmbH calibration laboratory for optical radiation measurements quantities. As with all light detectors supplied by Gigahertz-Optik calibration of absolute detector responsivity as well as detector individual measured relative spectral responsivity data is included.
|UV-A (320 - 400) nm Sensor||
|UV-BC (250 - 320) nm Sensor||
|UV-C 254 nm Sensor||
(5 - 40) °C
|Produktname||Product Image||Description||Show product|
|X1-4||Broadband radiometer for radiation protection measurements of UV radiation devices.
Features: Mobile ligh meter for UV radiation while working with artificial UV (UVSV). Erythema effective irradiance and detection of UV-C radiation in accordance with the UV protection Act for solaria. UVSV, DIN EN 60335-2-27 and DIN 5050-1:2010-01 conformity.